User manual OMRON E32-D82F DATASHEET
Lastmanuals offers a socially driven service of sharing, storing and searching manuals related to use of hardware and software : user guide, owner's manual, quick start guide, technical datasheets... DON'T FORGET : ALWAYS READ THE USER GUIDE BEFORE BUYING !!!
If this document matches the user guide, instructions manual or user manual, feature sets, schematics you are looking for, download it now. Lastmanuals provides you a fast and easy access to the user manual OMRON E32-D82F. We hope that this OMRON E32-D82F user guide will be useful to you.
Lastmanuals help download the user guide OMRON E32-D82F.
Manual abstract: user guide OMRON E32-D82FDATASHEET
Detailed instructions for use are in the User's Guide.
[. . . ] Fluid level sensor (contact type)
High-accuracy detection of fluid level in washing tank.
Uses Teflon (PFA) with excellent chemical and oil resistance. Capable of detecting high-temperature fluids such as sulfuric acid in a wafer washing tank. [. . . ] Using teaching without work Perform teaching with the tip of the fiber unit in the fluid. (The sensitivity is set to the top 10% of the received light intensity in fluid for stronger performance with respect to fluctuations in received light intensity due to fluid leakage, and thus teaching without work for high viscosity fluids is effective. ) Sensitivity control type
Sequence Detection state Sensitivity adjuster
2. Using teaching with/without work Perform teaching after the object has been removed from the fluid, and then repeat teaching with the object in the fluid. (Teaching with/without work is effective for fluids in which bubbles form at high temperature. )
Note: If set to the maximum sensitivity with the object removed from the fluid, detection of the fluid will no longer be possible.
Adjustment procedure Determine the position A at which the incident light indicator lamp (red) illuminates as the sensitivity control is gradually increased from the minimum setting after the object has been removed from the fluid. · If the red indicator lamp illuminates at the maximum sensitivity setting, gradually decrease the sensitivity control from the maximum setting with the object in the fluid, and determine the position B at which the incident light indicator lamp (red) goes off. · If the red indicator lamp goes off at the maximum sensitivity
Set the sensitivity control to C midway between A and B. At this time, verify that the stability indicator lamp (green) illuminates both with and without fluid.
Installation · Use the no-bending section to secure the fiber unit. If the fiber unit is secured without using the no-bending section, the fluid level detection position may shift. · Influences from the sides or bottom may interfere with detection. In that case, remove to a distance that is not subject to these influences, or apply a black coating to the sides and bottom. · If you need to use the system in a dangerous location, use only the fiber unit in the dangerous location and place the amplifier unit in a safe location. For adjustment About the fluid level detection position The fluid level detection position is located 5. 2±2 mm from the tip of the Surface level detection position Teflon section (see the 5. 2±2 mm (From a Teflon part tip) diagram at right). [. . . ] Freely cut because 2m part of optical fiber at amplifier side is made from a plastic fiber.
ALL DIMENSIONS SHOWN ARE IN MILLIMETERS. E41E-EN-01 In the interest of product improvement, specifications are subject to change without notice.
Advanced Photoelectric Sensors [. . . ]
DISCLAIMER TO DOWNLOAD THE USER GUIDE OMRON E32-D82F
Lastmanuals offers a socially driven service of sharing, storing and searching manuals related to use of hardware and software : user guide, owner's manual, quick start guide, technical datasheets...
In any way can't Lastmanuals be held responsible if the document you are looking for is not available, incomplete, in a different language than yours, or if the model or language do not match the description. Lastmanuals, for instance, does not offer a translation service.
Click on "Download the user Manual" at the end of this Contract if you accept its terms, the downloading of the manual OMRON E32-D82F will begin.