User manual OMRON E5 K-T DATASHEET
Lastmanuals offers a socially driven service of sharing, storing and searching manuals related to use of hardware and software : user guide, owner's manual, quick start guide, technical datasheets... DON'T FORGET : ALWAYS READ THE USER GUIDE BEFORE BUYING !!!
If this document matches the user guide, instructions manual or user manual, feature sets, schematics you are looking for, download it now. Lastmanuals provides you a fast and easy access to the user manual OMRON E5 K-T. We hope that this OMRON E5 K-T user guide will be useful to you.
Lastmanuals help download the user guide OMRON E5 K-T.
Manual abstract: user guide OMRON E5 K-TDATASHEET
Detailed instructions for use are in the User's Guide.
[. . . ] Press the Display Key to shift to the display for the time of step 0.
Step No. 0 1 2 3 50 100 100 50
Time (hours. minutes) 0. 00 0. 20 0. 40 0. 20
Time parameter for step 0. Press the Display Key to shift to the display for the number of steps.
Parameter for the number of steps used Number of steps (Default setting: 8) Pattern No. [. . . ] Operation Indicators · OUT1 Lights when the pulse output function assigned to control output 1 turns ON. · OUT2 Lights when the pulse output function assigned to control output 2 turns ON. · SUB1 Lights when the output function assigned to auxiliary output 1 turns ON. · SUB2 Lights when the output function assigned to auxiliary output 2 turns ON. Up Key/Down Key Press to increase or decrease the value on the No. 2 display.
Pattern Number Operation indicators
Up Key/Down Key Display Key RUN/RST Key
Digital Controller E5AK-T/E5EK-T
Advanced, Compact Programmable Digital Controllers Ideal for Worldwide Use
· Offers up to four patterns of simple programming control (16 steps per pattern). · 24 VAC/DC types are also available.
List of Models
Description Base Unit Model E5CK-TAA1 AC100-240 E5CK-TAA1-500 AC100-240 E5CK-TAA1 AC/DC24 E5CK-TAA1-500 AC/DC24 Note: A single Output Unit and Option Unit can be mounted to each Base Unit. Description Output Unit E53-R4R4 E53-Q4R4 E53-Q4HR4 E53-C4R4 E53-C4DR4 E53-V44R4 E53-Q4Q4 E53-Q4HQ4H Description Option Unit E53-CK01 E53-CK03 E53-CKB E53-CKF Model RS-232C RS-485 Event input: 1 point Transfer output (4 to 20 mA) Model Relay/Relay Pulse (NPN)/Relay Linear (4 to 20 mA)/Relay Linear (0 to 20 mA)/Relay Linear (0 to 10 V)/Relay Pulse (NPN)/Pulse (NPN) Pulse (PNP)/Pulse (PNP) Specification Temperature Controller Pulse (PNP)/Relay Specification Standard model Standard model with terminal cover Standard model Standard model with terminal cover Specification
The Digital Controller can be provided together with an inspection report. Refer to the following legend with the suffix "K" when ordering a model provided together with an inspection report. E5CK-TAA1-K
Accessories (Order Separately)
Name Terminal Cover E53-COV07 Model
Do not touch any of the terminals while the power is being supplied. If there is a large power-generating peripheral device and any of its lines near the Controller, attach a surge suppressor or noise filter to the device to stop the noise affecting the Controller system. In particular, motors, transformers, solenoids and magnetic coils have an inductance component, and therefore can generate very strong noise. When mounting a noise filter on the power supply to the Controller, be sure to first check the filter's voltage and current capacity, and then mount the filter as close as possible to the Controller. Use within the following temperature and humidity ranges: · Temperature: 10°C to 55°C (with no icing or condensation) Humidity: 35% to 85% (with no icing or condensation) If the Controller is installed inside a control board, the ambient temperature must be kept to under 55°C, including the temperature around the Controller. If the Controller is subjected to heat radiation, use a fan to cool the surface of the Controller to under 55°C. Store within the following temperature and humidity ranges: · Temperature: 25°C to 65°C (with no icing or condensation) Humidity: 35% to 85% (with no icing or condensation) Never place heavy objects on, or apply pressure to the Controller that may cause it to deform and deteriorate during use or storage. Avoid using the Controller in places near a radio, television set, or wireless installing. These devices can cause radio disturbances which adversely affect the performance of the Controller.
Be sure to observe these precautions to ensure safe use. · Do not use the product in places where explosive or flammable gases may be present. [. . . ] Mount the Unit horizontally.
To reduce inductive noise influence, the lead wires connecting the input type to the Digital Controller must be separated from the power lines and load lines. Use lead wires having a small resistance for platinum resistance thermometers.
Wire the terminals of the Unit using solderless terminals. The tightening torque applied to the terminal screws of the Unit must be approximately 0. 78 N·m or 8 kgf·cm. [. . . ]
DISCLAIMER TO DOWNLOAD THE USER GUIDE OMRON E5 K-T
Lastmanuals offers a socially driven service of sharing, storing and searching manuals related to use of hardware and software : user guide, owner's manual, quick start guide, technical datasheets...
In any way can't Lastmanuals be held responsible if the document you are looking for is not available, incomplete, in a different language than yours, or if the model or language do not match the description. Lastmanuals, for instance, does not offer a translation service.
Click on "Download the user Manual" at the end of this Contract if you accept its terms, the downloading of the manual OMRON E5 K-T will begin.